This helpful list contains terminology used frequently in the weighing industry, along with easy-to-understand explanations. New information is updated regularly, so make sure to visit often.
A weighing function that allows the device to total multiple weighing results.
These terms are often used interchangeably to indicate how well a scale displays the correct results. Accuracy is the ability to display a value that matches the ideal value for a known weight. See also: Resolution, Increment, Readability
- Ambient Conditions
The environmental conditions (temperature, humidity, pressure, temperature, etc.) of the area surrounding the load cell.
- Analytical Balance
A balance that weighs very small quantities in very tiny increments. Usually refers to sub-milligram weighing results.
- Animal Weighing
Some balances and scales feature a mode to allow accurate measurement of living animals that might move during weighing. The scale works out an average weight based on the force exerted by the animal over an extended period of time. Once the internal calculation is completed by the balance, it will hold the value. See dynamic weighing.
- Auto Zero Tracking
Display automatically zeroes to adjust for slow drift or changing environments.
- Backlit Display
The liquid crystal display (LCD) of some balances features a backlight to improve visibility of weighing results.
A weighing machine. The terms "scale" and "balance" are often incorrectly used interchangeably. A balance determines mass by balancing an unknown mass against a known mass, as with a two-pan assay balance. Modern balances usually are designed with a force restoration mechanism, which creates a force to balance the force due to the unknown mass. See scale. See also: Scale
- Battery Powered
The balance or scale are said to be battery powered when the units are operated using batteries. The battery powered units are portable and hence are often suitable for field applications.
- Baud Rate
The speed of communication when using the RS-232 interface. The greater the number the faster the data is sent between 2 devices. Usually balances use 300 to 9600 baud.
- Below Balance Weighing
To determine the specific gravity of a sample, below balance weighing is needed. The sample is tied to a wire attached to the under-hook of a balance and immersed into the liquid to know the weight of the sample in that liquid.
- Body Mass Index (BMI)
Measure of body fat that is based on adult height and weight. BMI is calculated by dividing a person's weight in kilograms by the square of their height in meters (kg/m2).Categories include underweight, normal, overweight and obese.
A procedure that enables the balance or scale to be reset to a known standard mass.
- Calibration Mass
A mass of known value that meets certain standards and is used to calibrate a balance to ensure accuracy.
The maximum weight that can be placed on the scale or balance.
A function that compares the current weight to a predetermined setting and indicates if the load is below, above, or within an acceptable range.
- Compact Balance
A small digital balance (usually portable) used for basic weighing tasks.
Change that takes place in load cell output while under load, and with all environmental conditions and other variables remaining constant. Creep usually is measured with rated load applied and expressed as a percent of rated output over a specific duration.
Balances can be programmed to display and print the date on which weighing operations were performed.
- Density Determination
Density is a measure of mass per unit volume. A balance is used to determine the density of solid or liquid materials. Density of a solid material is found by weighing it in air, and then suspending it in liquid and weighing it again. To determine the density of a liquid, a special float of known volume is weighed in air and then in the liquid. Density can be determined manually or automatically calculated by software in some balances.
- Digital Filters
Filters can help reduce the influence of draft or vibration, enabling the balance to display a stable reading more quickly.
The number of increments in weight that a scale offers. See increments.
Gradual change in the readout of a weighing instrument at a constant load.
- Dynamic Weighing
A mode featured on some balances and scales for weighing moving animals or measuring fluid samples. The device works out an average weight based on the force exerted by the animal or sample over an extended period of time. Once the internal calculation is completed by the balance, it will hold the value. See Animal Weighing.
- EC Approved
EC (European commission) approvals are required in the UK and Europe for weighing scales for buying and selling goods by weight.
- Eccentric Loading
See Off-centre loading. See also: Off-centre Loading
- External Calibration
The calibration process of a balance using a known external weight (often referred to as "suggested calibration weight").
- Force Restoration
A method to determine an unknown mass by balancing the force produced by the mass with a system-created force using coils mounted in a magnetic field. Also called "servomotor" or "force motor."
Formulation, also referred to as the recipe calculation function, is a selectable weighing mode on precision and analytical balances that allows the user to easily store, add and calculate the weight of multiple ingredients in order to provide a total weight value.
Good Laboratory Practice (GLP) generally refers to a system of management controls for laboratories and research organisations to ensure the consistency and reliability of results. Laboratory balances can provide GLP printouts with traceability, which include date, time, balance identification number, etc.
- GLP/GMP Printout Capability
Program that provides a specific set of information on the balance and the sample weight, in conformance with Good Laboratory/Manufacturing Practices.
Device on which weighing or counting results are displayed. The scale or balance often can be programmed for a variety of applications and weighing units using software accessed via the indicator.
- Ingress Protection (IP) Rating
The IP rating classifies the degree of protection against the intrusion of solid objects such as dust and water. It provides specific information about a scale's washdown ability, or if it can be used in a wet or dusty environment.
- Internal Calibration
An automatic calibration process that uses a weight available inside the balance. Calibration can be programmed by the user to occur when the temperature changes, at regular intervals, or when needed.
- Internal Rechargeable Battery
Digital balances are often powered by an internal battery that is recharged when connected to the main power supply.
This procedure uses adjustable feet and a levelling bubble to ensure the balance's platform is horizontal. Levelling helps the balance adapt to uneven or sloped surfaces, ensuring repeatable and accurate results if the balance is moved.
The ability of a scale or balance to show the correct value throughout the weighing range. Linearity can be tested by placing known weights on the balance from near zero to full capacity.
- Load Cells
A load cell is a specialist transducer that is used to convert force (weight) exerted onto it into measurable electrical output. Load cells are used in many types of industrial machinery and equipment, however they are most often used in weighing scales. See also: Strain Gauge Load Cells
Mass is a physical property of a material that provides a weight when gravity pulls the material toward the Earth. The units of mass are gram and kilogram. The terms "mass" and "weight" often are used interchangeably.
A set of procedures that can be followed to change the characteristics of a balance. The menu consists of a number of options (parameters) that are set at the factory or can be specified by the user.
- Moisture Analyser/Moisture Balance
A weighing device that quickly measures samples before and after drying to determine moisture levels. Suitable for a variety of industries, moisture analysers are used in product development labs to test foods and pharmaceuticals, or to assess the moisture content in sludge during the wastewater treatment process.
- Moisture Determination
Balance function that calculates moisture percentage by comparing an initial sample weight to the same sample’s weight after drying.
- Multiple Weighing Units
In reference to a weighing scale or balance that has more than one weighing unit. See Weighing units. See also: Weighing Units
- Net Total Formulation
Balance function that automatically adds the individual weights of several components and displays the total.
- Net Weight
Weight of the load on the pan minus the container (tare) weight.
The National Type Evaluation Program (NTEP) sometimes referred to as NIST, is an evaluation procedure used in the USA for certifying that a scale or balance is legally approved for buying or selling goods by weight.
- Off-centre Loading
The ability to display the same value when a weight is placed anywhere on the weighing platform. Tested by weighing in the centre of four quadrants on the platform. Also called "eccentric loading" or "shift test."
OIML (International Organization of Legal Metrology) is a regulatory body that develops model regulations and standards for use by legal metrology authorities and the weighing industry.
A set of procedures that can be followed to change the characteristics of a balance. The menu is made up of a number of options, called parameters, that can be set at the factory or in some cases by the user. See also: Menu
- Parts Counting
A weighing application that uses an average piece weight (representative sample) to determine the number of items having the same mass by placing them on the platform. Piece weight is entered by the user or determined by measuring a sample on the scale.
- Percent Weighing
Using a balance to determine the weight of an item as a percentage of a reference weight.
- Precision Balance
A balance used to weigh quantities to a very precise number, usually up to one milligram. Precision balances come in a wide variety of capacities and pan sizes.
The degree of agreement between repeated measurements of the same mass on the same balance, under the same conditions.
See Repeatability. See also: Repeatability
A method of transmitting data via wires, allowing balances to communicate with printers or computers. Scales and balances with data communication capabilities use RS-232 interfaces to connect to peripheral devices.
A weighing machine. The terms "scale" and "balance" are often incorrectly used interchangeably. Historically, a scale was a device that displayed weight by measuring a deflection, such as a spring scale. Modern scales generally use springs. Weighing devices with strain gauge load cells are sometimes referred to as scales. See Balance. See also: Balance
- Security Bracket
A built-in bracket that attaches the balance to a lock and cable to help prevent theft or relocation of the device.
- Shift Test
See Off-centre loading and Eccentric loading. See also: Off-centre Loading
- Specific Gravity
Specific gravity is a dimensionless quantity. It is the ratio of density of a substance to the density of reference substance ( water, air, alcohol, etc.). See Density Determination. See also: Density Determination
The point at which results do not change after a weight has been added to the platform.
- Stable Indicator
A feature or symbol that appears on the display indicating that the balance has provided results and the displayed value no longer will change.
- Strain Gauge Load Cells
A method to determine mass using a mechanical component that is deflected by the addition of an unknown mass. The amount of deflection is measured by resistive elements (strain gauges) mounted on the mechanical component.
Tare means setting a display to show zero weight. This is used to remove the weight of any packing or containers, leaving only the weight of the material on the display. The tare value is subtracted from the remaining weight added to a scale. For example, if 200g is tared on a scale with a capacity of 500g, a capacity of 300g remains. "Tare" and "zero" are frequently used interchangeably.
- Tare Weight
The weight of a container or package that is excluded when weighing a sample.
- Temperature Compensation
The ability to correct errors in the weighing system due to temperature changes, generally achieved via software or electronics on precision balances.
Balances can be programmed to display and print the time at which weighing operations were performed.
The ability of a balance to add together multiple results when weighing separate items.
- Triple Beam Balance
A mechanical balance that uses counter weights to determine an object's mass.
To find the weight (mass) of an unknown against a standard known mass.
- Weighing Units
Unit measure displayed on a balance that provides the weight of an unknown mass, such as grams, kilograms, milligrams, pounds, ounces, carats, grains, Newtons, etc.
The weight of an object is the result of gravity pulling a mass toward the Earth. When a balance is calibrated using a known mass, any unknown mass placed on the scale will have a weight proportional to the known mass. A weight can also be any mass that is placed on the scale.
This sets the display to show zero weight. Used to reset the zero condition of a scale when small amounts of material are on the platform. Zero does not take away from the capacity of a scale; however, it works only in a very small range around the original zero condition for the scale. Zero and tare sometimes are located on the same key, as the terms often are used interchangeably.